- *Corresponding Author:
- B. K. Jayanna
Department of Chemistry, B. N. M. Institute of Technology, Bengaluru-560 070, India
E-mail: [email protected]
|Date of Submission||03 September 2013|
|Date of Revision||07 August 2014|
|Date of Acceptance||12 August 2014|
|Indian J Pharm Sci 2014;76(5):452-455|
A simple, rapid and highly sensitive spectrophotometric method is developed for the determination of risperidone in tablet formulation. The method is based on the oxidation of drug using potassium permanganate in alkaline medium and excess potassium permanganate oxidizes 1,10-phenanthroline Fe(II). The measurement of decrease in absorbance of 1,10-phenanthroline Fe (II) was done at 415 nm. The beer's law is obeyed in the concentration range of 5.0 to 40.0 μg/ml and molar absorptivity is found to be 7.3932 × 10 4 l/mol/cm. The proposed method is well suited for the pharmaceutical formulations.
Risperidone, potassium permanganate, 1,10-phenanthroline Fe (II), oxidation, tablets, spectrophotometry
The commonly used analytical techniques for the determination of risperidone are differential pulse polarography  chemiluminescence assay , HPLC-MS/MS , chiral chromatography , LC-MS , LC-tandem mass spectrometry , LC with electrochemical detection [7,8] LC with UV detection , LC with coulometric detection  and visible spectrophotometric method . Colorimetric determination of drugs using the ferric chloride assay is simple and rapid. However it offers relatively low resolution and cannot be used to determine total drug concentrations. The reddish orange color of 1,10-phenanthroline Fe (II) complex  is extremely stable in acidic as well as in alkaline medium. Hence it is used in the present method. The aim of the present work is to develop simple method for the determination of risperidone in different dosage form. The proposed method stands atop over the reported method with respect to sensitivity and the method neither requires extraction nor prior separation of the drug.
Risperidone was received from Max India’s Pharmaceutical Divison, Nanjanagud, Mysore. Tablets Risperdal (risperidone; 2 mg/tablet) of Johnson and Johnson Limited, Mumbai, India and Rispond (risperidone; 3, 2, and 1 mg/tablet) of Micro Labs Limited, Bangalore, India were purchased from local market. Potassium permanganate and anhydrous potassium carbonate of Glaxo Ltd., Mumbai, India, 1,10-phenanthroline monohydrate of E. Merck, India and ferrous sulphate heptahydrate of S. D. Fine Chem., Mumbai, India were procured and were of analytical grade. UV/Vis spectrophotometer with 10 mm matched quartz cells were used for absorbance measurement
Standard stock solution of risperidone (100 μg/ml) was prepared by dissolving 10 mg of risperidone in distilled water and diluted to 100 ml. Potassium permanganate (0.0012 mol/l) and potassium carbonate (0.1 mol/l) were prepared by dissolving 20 mg of potassium permanganate and 1383 mg of potassium carbonate, respectively in distilled water and diluted to 100 ml. 1,10-Phenanthroline Fe (II) sulphate prepared by dissolving 1500 mg of 1,10-phenanthroline monohydrate and 700 mg of ferrous sulphate in 100 ml water. 10 ml of this solution was diluted to 100 ml.
Different aliquots of the standard risperidone (05 to 40 μg/ml) were transferred into a series of 10 ml volumetric flasks. Warmed on a water bath for about 5 min and to each one of these flasks 1.0 ml potassium permanganate (0.0012 M) followed by 1.0 ml of potassium carbonate were added. Again warmed on a water bath for about 1 min and 0.5 ml of 1,10-phenanthroline Fe (II) was added. The volume was made up to 10 ml with water and mixed thoroughly. Absorbance of these solutions were measured at 415 nm making zero absorbance with distilled water.
The calibration curve was prepared by recommended procedure as shown in fig. 1. Linear relationships between absorbance and concentration held over range of 05 to 40 μg/ml and other parameters are given in Table 1. The high valve of the correlation coefficient close to unity corroborates the linearity of the calibration plot. The high sensitivity of the method was indicated by the relatively high valve of molar absorptivity and low values of Sandell’s sensitivity.
|Beer’s limit (μg/ml) 10‑55||05‑40|
|Molar absorptivity (lmol−1cm−1)||7.3932x104|
|Sandel sensitivity (μg cm−2/0.001A)||0.005521|
Table 1: Optical characteristics of the proposed procedure
Precision of the proposed method was described by performing repeatability within the day (intraday precision) and intermediate precision on different days (interday precision) in three replicates. Repeatability and intermediate precision was performed for three different concentrations 10, 20 and 30 μg. The amount, relative error percentage (% RE) and relative standard deviation (% RSD) were found to be 9.98, 0.2, 0.86, 20.05, 0.25, 1.45 and 30.15, 0.5, 1.23 and 10.05, 0.5, 1.57, 19.98, 0.1, 0.65 and 30.12, 0.4, 1.98, respectively. The low values of the relative standard deviation percentage and relative error percentage specify the high precision and the good accuracy of the proposed method.
The robustness of the method was estimated by making small incremental changes in volume of potassium carbonate and effect of change was studied on the absorbance of the colored system. Alter has no influence on results. Further robustness of the method is that it does not involve extraction. The Ruggedness of the method was illustrated for the same concentration 10 μg under instrument I (UV Spectrophotometer Model-117, Systronics) and instrument II (UV Spectrophotometer Model-1800, Hitachi) and by two analysts in the same laboratory. Amount and relative standard deviation in both instances were found to be 9.99, 0.55, 10.02, 1.76 and 9.96, 1.65, 9.98, 0.86. The low intermediate precision valves (% RSD) in both instances indicating ruggedness of the method.
The effect of excipients was examined by determining 10 μg/ml of risperidone in presence of different excipients such as starch, glucose, cellulose, lactose and talc at different concentrations present in tablet formulation. The results specify that there is no interference from any excipients because redox reaction taxes place in weak alkaline medium. For the analysis of the formulation ten tablets of risperidone were weighed and finally powdered. The powder equivalent to 10 mg was weighed accurately and dissolved in water. The residue was filtered into 100 ml volumetric flask. The residue was washed repeatedly with distilled water and washings were added to the filtrate. Final volume of filtrate was made up to the mark with distilled water and was analyzed according to the recommended procedure. The amount and relative standard deviation (%RSD) for the formulations Risperdal of 2 mg and Rispond of 1 mg, 2 mg and 3 mg were found to be 2.01, 0.18 and 1.04, 0.52, 2.05, 0.68 and 2.98, 0.3, respectively.
The method is based on the oxidation of risperidone with potassium permanganate in alkaline medium. The color of 1,10-phenanthroline Fe (II) decreases quantatively in the presence of excess potassium permanganate and oxidized drug due to oxidation of Fe (II) to Fe (III). The possible reaction scheme of the method is shown in fig. 2. The color is stable for 5 h for the concentration of drug in the range of 5-20 μg/ml and 1 h in the range of 25- 40 μg/ml. It was reported in the previous work  that absorbance was measured after 30 min of preparation of standard drug solution using potassium permanganate. In the present method absorbance was measured after the preparation of standard drug solution. The developed method is found to be simple, sensitive and accurate. The excipients present in the tablet dosage form did not hinder with the determination of risperidone. Hence, method can be used successfully for the determination of risperidone in tablet formulation.
The authors thank VTU, Belgaum, for providing the fund in doing research work.
- Christine J, Ravin J, Arun PJ. Differential pulse polarographic studies of risperidone in pharmaceutical formulations. CroaticaChemicaActa2006;79:541-4.
- Song Z, Wang C. Sensitive chemiluminescence assay for risperidone in pharmaceutical preparations. J Pharm Biomed Anal 2004;36:491-4.
- Cabovska B, Cox SL, Vinks AA. Determination of risperidone and enantiomers of 9-hydroxyrisperidone in plasma by LCMS/ MS. J Chromatog B AnalytTechnol Biomed Life Sci 2007;852:497-504.
- Danel C, Odou P, Barthelemy C, Azarzar D, Robert H, Bonte JP, et al. Analytical and semipreparativeenantioresolution of 9-hydroxyrisperidone, main metabolite of risperidone using high performance liquid chromatography and capillary zone electrophoresis. Validation and determination of enantiomeric purity. J Chromatogr A 2007;1163:226-36.
- Huang MZ, ShentuJZ, Chen JC, Liu J, Zhou HL. Determination of risperidone in human plasma by HPLC-MS/MS and its application to a pharmacokinetic study in Chinese volunteers. J Zhejiang Univ Sci B 2008;9:114-20.
- Bhatt J, Subbaiah G, Singh S. Liquid chromatography/tandem mass spectrometry method for simultaneous determination of risperidone and its active metabolite 9-hydroxyrisperidone in human plasma. Rapid Commun Mass Spectrom 2006;20:2109-14.
- Le Moing JP, Edouard S, LevronJC. Determination of risperidone and 9-hydroxyrisperidone in human plasma by high-performance liquid chromatography with electrochemical detection. J Chromatogr 1993;614:333-9.
- Locatelli I, Mrhar A, Grabnar I. Simultaneous determination of risperidone and 9-hydroxyrisperidone enantiomers in human blood plasma by liquid chromatography with electrochemical detection. J Pharm Biomed Anal 2009;50:905-10.
- Avenoso A, Facciola G, Salemi M, Spina E. Determination of risperidone and its major metabolite 9-hydroxyrisperidone in human plasma by reversed-phase liquid chromatography with ultraviolet detection. J Chromatogr B Biomed Sci Appl 2000;746:173-81.
- Schatz DS. Saria A. Simultaneous determination of paroxetine, risperidone and 9-hydroxyrisperidone in human plasma by highperformance liquid chromatography with coulometric detection. Pharmacology 2000;60:51-6.
- Narayana B, Divya NS, Development and validation of spectrophotomtric method for the determination of risperidone. Der Pharmacia Lettre 2011;3:104-9.
- Harris DC. Determination of Iron with 1,10-phenanthroline, line in: Quantitative chemical analysis. 6th ed. New York: Freeman WH and Company; 2003. p. 258-61.
- JayannaBK, Nagendrappa G, Arunkumar, Gowda N. Spectrophotometric estimation of Azithromycin in tablets. Indian J Pharm Sci 2012;74:365-7.